» The Miris Check solution is used for zero setting the Miris Human Milk Analyzer and control that cleaning has been done successfully
» Application: Only for validation of the preset internal calibration of the Miris HMA.
» Storage: Store the concentrate at room temperature (20-30°C) in a sealed container. Store the diluted solution in glass or plastic containers, out of direct sunlight. Keep away from foodstuff and animal feed.
» Instructions: Dilute 10 ml concentrate (1 tube) with 90 ml distilled or deionized water and mix. Warm to 40°C prior to use. The estimated time for preparation is 15 minutes for mixing the ingredients and to warm the solution to 40°C. The solution is to be used when preparing the instrument for analysis and also after cleaning if analysis is continued.
» 5 tubes per order of 10ml each
Miris is a company who specialize in human milk analysis of pre-term babies. They believe by knowing the macro nutrient composition of human milk, it can be determined whether milk fortification is needed.
Miris is operating on all continents and HMA™ is registered as a Medical Device in Europe, Japan and Indonesia.
Click here to watch a short video on how to mix the check solution. To follow the check procedure, click here.